You are using the RFmx GSM toolkit obtain results for ModAcc, ORFS, and PVT;
When you generate an uplink signal with only Slot 1/2 On， but two slots have different modulation data ，refer to below setting
Then open the RFmx GSM toolkit to measure the signal ，we find the result as below：
You could see the RFmx GSM only support measure one slot，above picture show that it was measuring the Slot1. Is there one solution to measure slot2 ？
The trigger default setting of RFmx GSM was IQ Power Edge trigger and the measure length was 1 slot . So we could see it only measured the Slot1 like above picture.
1.Change the trigger type as Digital trigger, you will see below result:
There is no real signal data in the measurement result , because it was measuring the slot0 which was un-enabled.
2. Change the delay setting as 576.2us , we could see the same result when measuring with power edge trigger
3.Change the delay as 576.2*2=1152.4us, and change the Modulation Type as"8PSK" , we could see the measurement result for Slot2
We could see the measurement result for each slot by setting the delay value with digital trigger.
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.