If your network admin imposes specific addressing requirements or if you are having trouble with a link-local connection between your Ethernet cDAQ device and your computer, you may need to configure a static IP address on your cDAQ device.
The factory reset the Ethernet cDAQ device by holding down the reset button for more than 5 seconds. This will put the device into a default state and configuration. See the manual for the device-specific default configuration.
Connect the Ethernet cDAQ device to the Ethernet port on your computer or the network you would like to connect the device to.
In Measurement & Automation Explorer (MAX), expand My System»Devices and Interfaces. Right-click Network Devices, then Find Network NI-DAQmx Devices.
Figure 1. Find Network NI-DAQmx Devices
In the Available Devices list, select your Ethernet cDAQ device, then click Add Selected Devices.
Figure 2. Add Selected Devices
On the Network Settings tab (1), choose Static in the Configure IPv4 Address dropdown (2), and enter a local IP address (e.g. 192.168.2.10) in the IPv4 Address field. Click Save to apply the settings. You will lose connection while the device restarts.
Figure 3. Network Settings
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.