This content is not available in your preferred language.
The content is shown in another available language. Your browser may include features that can help translate the text.
This document explains how to install the application and NI-ELVISmx driver software and set up your myDAQ Student Data Acquisition Device. This page also contains links to several resources, such as software download pages, installation troubleshooting, and measurement examples.
When you open the box, you'll see the following items:
The NI myDAQ Quick Start Guide . It contains additional information about various device connections not found in this document, as well as a quick measurement example to get you up and running.
A tray containing the following items:
NI myDAQ Student Data Acquisition Device
NI myDAQ Software Suite DVD
Digital Multimeter (DMM) Probes
3.5 mm Audio Cable
Figure 1. NI myDAQ Quick Start Guide
Figure 2. Contents of NI myDAQ Box
Keep the tray and box to store your NI myDAQ and accessories. You also can store a breadboard and other accessories on the right side of the tray. You'll need all these items to get set up.
You can install the necessary software from the media that shipped with your myDAQ, or from ni.com/downloads.
If you are using the software installation media—Simply insert the media in your computer and follow the onscreen instructions. Note that the installation process may take from 20 to 40 minutes.
If you are downloading your software—Go to the following pages to download the necessary installers. Be sure to install application software, such as LabVIEW, first, then install your ELVISmx driver software.
LabVIEW Development System for Windows
Circuit Design Suite Software (includes NI Multisim, Ultiboard)
NI ELVISmx Driver Software (NI myDAQ hardware driver )
Note: Using NI ELVIS, too? For software version and operating system support for the NI ELVISmx hardware driver, consult the NI ELVISmx Readme, available on the software media shipped with your device or on the NI ELVISmx software download page.
After the software installation is complete, you can set up your NI myDAQ hardware.
Connect one end of the USB cable to your NI myDAQ and the other end to the computer.
Confirm the blue LED lights up to indicate that the device is being powered.
If the light does not illuminate, refer to the Troubleshoot your NI myDAQ article for more information on possible causes and solutions.
The ELVISmx offers several virtual instruments, which are computer-based versions of some of the more commonly used laboratory instruments. NI ELVISmx is also shared with NI ELVIS , a prototyping platform for circuits and electronics.
The ELVISmx Instrument Launcher helps you get started taking measurements easily. It provides some of the most common functions you can perform with your myDAQ.The ELVISmx Instrument Launcher should automatically appear after you connect your hardware. If the ELVISmx Instrument Launcher does not appear, refer to the Troubleshoot your NI myDAQ article for more information on possible causes and solutions. You can also manually launch the ELVISmx Instrument Launcher by selecting Start»All Programs»National Instruments»NI ELVISmx for NI ELVIS & NI myDAQ»NI ELVISmx Instrument Launcher.
Now that you have installed your software and set up your NI myDAQ, you are ready to take your first measurement using NI myDAQ and NI ELVISmx instruments .
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.