# Calculating the Absolute Accuracy or Uncertainty of NI Current Device

Updated Apr 25, 2019

## Reported In

### Hardware

• NI-9208
• C Series Current Input Module
• C Series Current Output Module

## Issue Details

I want to calculate the standard uncertainty of my NI current Device and incorporate this into the total uncertainty for my current measurement tests.
I have tried using the article Should The Absolute Accuracy Of A Measurement See A Constant or Dynamic Offset and the white paper Calculating Absolute Accuracy or System Accuracy.

However, the equations use voltage and I want to calculate the accuracy for my current devices/measurements.

Absolute Accuracy =± [(VoltageReading x GainError) + (VoltageRange x OffsetError) + NoiseUncertainty]

## Solution

The equation in the articles mentioned above can still be used for current measurements. Wherever voltage is mentioned, replace this with current.
As an example, see the absolute accuracy or standard uncertainty calculation for the NI-9208 Current Input module below.

In the manual, NI 9208 Datasheet , the Accuracy data is displayed in the Specifications section, in a table. Let's assume a measured value of 20 mA for this example.

To calculate the gain error, we multiply the Percentage of Reading, here ± 0.76 %, by the measured value, in SI units as below. (VoltageRange x OffsetError)
Then calculate the offset error by multiplying the range, here 22 mA (from the footnote), by the Percentage of Range, ± 0.04 %, as below. Adding these two together will give ± 0.0001608 A or 160.8 µA. This is the absolute accuracy of the current module.

For your complete system accuracy, you will need to include all component absolute accuracy values, as shown in the equation below. 