My SignalExpress project shows Error -200088 and I'm unable to acquire any data. If I try to configure my acquisition step, I can't find any devices or channels. SignalExpress is not showing measurements anymore.Error -200088: Task specified is invalid or does not exist.
The NI-DAQmx driver can get corrupted for multiple reasons. Error -200088 refers to the software not being able to find the task in the NI MAX database, which can relate to: the database being corrupted, or the drivers are missing. If you changed to another computer and you just installed SignalExpress, it's possible you're missing the NI-DAQmx driver to work with your National Instruments data acquisition devices.NI introduced DAQExpress and FlexLogger™ for the new generation of Data Acquisition software. It's highly recommended to consider them for future applications.
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful