My SignalExpress project shows Error -200088 and I'm unable to acquire any data. If I try to configure my acquisition step, I can't find any devices or channels. SignalExpress is not showing measurements anymore.Error -200088: Task specified is invalid or does not exist.
The NI-DAQmx driver can get corrupted for multiple reasons. Error -200088 refers to the software not being able to find the task in the NI MAX database, which can relate to: the database being corrupted, or the drivers are missing. If you changed to another computer and you just installed SignalExpress, it's possible you're missing the NI-DAQmx driver to work with your National Instruments data acquisition devices.NI introduced DAQExpress and FlexLogger™ for the new generation of Data Acquisition software. It's highly recommended to consider them for future applications.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.