Expand NI-DAQmx Devices from the configuration tree. Select your device and click Test Panels on the toolbar, as shown in figure below.From the test panel of your device, click the Analog Output tab. In Channel Name select a device supported analog output channel, enter an Output Value, and click Update. The step is shown in figure below.Without closing the test panel window, select Analog Input. In Channel Name select the same analog output channel and its corresponding ground. Click Start and the chart should begin updating with the value entered from the Analog Output tab. The step is shown in figure below.
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.