This error is triggered by the Sense Lead Error Detection
feature of the PXIe-4135, PXIe-4136 and PXIe-4137 when configured to use Remote Sense
. This feature is to alert you when there may be an issue with the connection to your DUT.
- Verify that the HI/LO and HI SENSE/LO SENSE lines are properly and firmly connected to the SMU and the DUT
- Measure the voltage across HI & HI SENSE and LO & LO SENSE lines, respectively, to make sure they are not greater that 3 V.
Once narrowing down issue, reset the device to clear the error and confirm you are following SMU Best Practices