Firstly, follow the steps outlined in What is the Process For Resetting the MAX Database? to see if a corrupt database was the cause of the issue.If the issue persists after these steps, it is likely you have a bad installation of MAX. Therefore, we will uninstall MAX and then reinstall it.
Steps to uninstall MAX:
Open the control panel by selecting Start » Settings » Control Panel
Double-click Add/Remove Programs
Select National Instruments Software and click the Change/Remove button
Select NI Measurement & Automation Explorer from the list of National Instrments software and click the Remove button
When asked if you are sure if you want to remove all components, select Remove All
Steps to reinstall MAX:
MAX is installed when any National Instruments driver is installed.
Uninstall a driver (such as NI-DAQmx, NI-488.2, NI-VISA) on your system by following steps 1-3 from above
Reinstall the current version of the driver which will then also install MAX
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.