I am using a combination of XNET and DAQ C Series cards in a TSN enabled network with CompactDAQ. Can I synchronize both tasks to start acquiring data at the same time? Moreover, will the XNET message timestamps and DAQ timestamps be synchronized by default?
The DAQmx and XNET modules within one chassis should already share a timebase, which is in turn synchronized to the timebases of other TSN devices on the network.
You will need to trigger the XNET session appropriately. A good place to start is the "Synchronize C Series CAN with DAQmx Analog Input" shipping example (Help > Find Examples > Hardware Input and Output > CAN > NI-XNET > Synchronization).
An important thing to note is that the trigger routed between the DAQmx task and XNET session is a digital signal on the chassis backplane and is not sent over the network. This means that a DAQmx task must exist on each chassis that you are using an XNET module in. Otherwise, the XNET session cannot be triggered.
You can find more information about the TSN features, and ask questions about the implementation of the APIs, in the NI's discussion forums: LabVIEW Time Sensitive Networking (TSN)
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.