The update rate of frequency task is the lower limit of the channel frequency. If this is set to 1Hz then the update rate will be 1Hz. The way that the frequency task is working in VeriStand is that the VeriStand engine calculates the frequency by counting the number of pulses between ticks of the Primary Control Loop. Thus, the accuracy of this calculation will be tied directly to the jitter of the PCL. For example, if you measure the PCL and determine a 5% deviation, we can expect a 5% error in the frequency measurement. This method is software timed and because of this it is subject to bad performance.It is recommended to use a Pulse Measurement task instead. This task is an option if using one of our X series devices. Another option is to use an R series device and implement the frequency measurement on the FPGA.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.