The Current Programming Accuracy/Resolution section of the PXIe-4110 Specification Sheet makes reference to a current level and a current limit. What should I use?
The current level setting is applicable only if the output function of the channel is set to NIDCPOWER_VAL_DC_CURRENT. It is the current the device attempts to generate.
The current limit setting is applicable only if the output function of the channel is set to NIDCPOWER_VAL_DC_VOLTAGE. It is the current that the output should not exceed when generating voltage.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.