I read some values with a PXI DMM multimeter 4080 and then I have a PXI-multiplexer 2575. If I have channels 3, 4 and 5 connected, there is a shift in values : I read the value of channel 3 on channel 4. What can I do to avoid this behavior ?
To avoid this behavior, you must be careful to configure all triggers correctly . On the example of LabVIEW, there is the project "Switch - Scanning with a DMM - Handshaking" which shows that you have to configure the triggers three times. You can see the triggers configuration at A2, B3 and B5 This is the snippet of this example.:
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.