I configured a Analog Input multidevice task
in order to take advantage of channel expansion
, by adding channels from different PXI devices. Since one of the device (PXIe-4339) is used for bridge input, I added DAQmx Perform Bridge Offset Nulling Calibration (VI)
in my LabVIEW code (or DAQmxPerformBridgeOffsetNullingCalEx
in my C or LabWindows/CVI code).
However, when running my program, I get the following error:Error -201444 occurred at DAQmx Perform Bridge Offset Nulling Calibration.vi:1900001
Multidevice tasks cannot use the on-demand sample timing type. Configure timing to synchronize and acquire samples from multiple devices.
The error is unexpected since I correctly configured the sample clock to obtain a hardware-timed task (continuous sampling).
How can I fix the error?