When I run the example "Strain (Rosette) - SW-Timed Input" in NI 9236, I get Error -201087: Task contains physical channels on one or more devices that require you to specify the Sample Clock rate. Use the Sample Clock Timing function/VI to speficy a Sample Clock rate.
The example does not work right away because NI 9235/36/37 does not support software timing, so it is necessary to add a sample clock to the task using DAQmx Timing VI as follows:
This error may occur in other devices not listed in this article. If it is your case try this solution.If you are getting incorrect measurements, try setting "Do Offset Null?" to true to calibrate your device.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.