I noticed that my cDAQ temperature input module, such as the NI-9213, is able to sample at rates faster than specified in the datasheet. However, if I do this on PXI temperature input modules, such as the PXIe-4353, I get error 4730001: Requested value is not a supported value for this property. The property value may be invalid because it conflicts with another property. Why can't my PXI module sample as fast as my cDAQ module?
This behavior is only due to slight differences in how the DAQmx driver assigns sample rates. The driver doesn't prevent clock speeds that are out of spec in the cDAQ line, although it does prevent these speeds on the PXI platform.Be aware that, even though the cDAQ task allows you to choose a higher sample rate, you are not actually obtaining a higher resolution signal. The limitation mentioned in the specifications sheet is still correct - any faster sample rate will just cause the same data point to be sampled multiple times.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.