I am using an NI-9215 module to measure the voltage signals and see the input data on a waveform chart. Seeing the data, it seems the signals has some offset and noise.
There is a possibility to solve the issue by rewiring like expalining below:
Differential Voltage Signal
For grounded differential signals, connect the positive voltage signal to AI+ and the negative signal to AI–. For the NI USB-9215 with screw terminal, connect the signal reference to the COM terminal.
For floating differential signals, the voltage source may go outside of the common-mode signal range of the NI USB-9215 with screw terminal. If the voltage source is outside of the common-mode range, data read by the NI USB-9215 is not accurate. To keep the voltage source within the common-mode range, connect the negative lead of the signal to COM through a 1 MΩ resistor. The NI USB-9215A with BNC has internal circuitry that keeps the voltage source within the common-mode range.
Single-Ended Voltage Signal
Connect the positive voltage signal to AI+. Connect the ground signal to AI–. For the NI USB-9215 with screw terminal, you must also connect the ground signal to the COM terminal to keep the common-mode voltage in the specified range.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.