This content is not available in your preferred language.
The content is shown in another available language. Your browser may include features that can help translate the text.
I am trying to communicate with my instrument from LabVIEW and I receive the following error message:Error -1073807304 occurred at VISA Write
In order to fix this error, make sure that you are wiring in the correct VISA address for the instrument with which you are trying to communicate. In the case of GPIB devices, you can find the address in Measurement & Automation Explorer as in the following screenshot:
For example, in the above screenshot, the address should be GPIB0::0::INSTROther Troubleshooting Steps:
This usually means there are no listeners at the VISA address specified by the address string. It typically occurs when you specify an incorrect address (usually a GPIB address) to the VISA Open VI.
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.