Note that pulse measurements are only supported by NI X-Series and 2nd generation NI CompactDAQ devices. This error does not apply to the NI 9361 module.
For LabVIEW 2011 and prior examples, you can go to Help » Find Examples...
to open up the Example Finder. Then navigate to Hardware Input and Output » DAQmx » Counter Measurements » Period or Pulse Width
. For Sample Clocked Timing, choose Meas Pulse Width-Buffered-Sample Clock-Cont
. For Implicit Timing, choose Meas Pulse Width-Buffered-Cont.
For LabVIEW 2012 and later examples, open up the Example Finder and navigate to Hardware Input and Output » DAQmx » Counter Input » Counter - Read Pulse Width and Frequency (Continuous).vi
. To switch from Implicit to Sample Clock timing, just change the polymorphic Timing VI
to Sample Clock
and specify a Clock Source and a Rate.
Please reference Chapter 7 - Counters
of the X Series User Manual
for further references on pulse measurements.