There are a wide variety of reasons that this error can occur. Potential causes and corresponding solutions for Error -88705 or -88708 are stated below:
The NI Device Loader Windows Service (nidevldu) and/or NI Configuration Manager Service is not running, please start these services as described here. If the NI Device Loader still does not start automatically after restarting computer, you can try to uninstall all antivirus software in your computer and restart again.
This error is sometimes reported if the computer has not been restarted after the installation of an NI driver. Please try restarting your computer.
During a driver installation, firmware updates download with the driver for many NI devices. If you are using a USB device, this particular cause can be fixed by unplugging and re-plugging the device.
Occasionally, the NI-MAX database can become corrupted due to a software install/update or other events impacting the registry files owned by MAX. Reset the database, restart your computer and devices, and try again.
Check which NI drivers are needed for your device (you can find this in the user manual for your device), then Uninstall or Repair National Instruments Software or Drivers in Windows and reinstall the drivers from ni.com.
Windows may have associated the wrong driver with your device. You can manually set the driver that Windows associates with your device.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.