Potential causes and corresponding solutions for Error -88705 are stated below:
The NI Device Loader Windows Service (nidevldu) and/or NI Configuration Manager Service is not running, please start these services. If the NI Device Loader still does not start automatically after restarting computer, you can try to uninstall all antivirus software in your computer and restart again.
This error is sometimes reported if the computer has not been restarted after the installation of the DAQmx driver. Please try restarting your computer.
During a driver installation, firmware updates download with the driver for many NI DAQ devices. If you are using a USB devices, this problem can be fixed by unplugging and re-plugging the device.
MAX configuration database might be constantly throwing this error. Reset the database and restart your computer and devices and try again.
Check which drivers are necessary for your device. You can Uninstall or Repair National Instruments Software or Drivers in Windows, then reinstall the drivers normally. Drivers are available here.
Check from Windows Device Manager whether a third-party driver is set as prioritized over NI drivers.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.