This content is not available in your preferred language.
The content is shown in another available language. Your browser may include features that can help translate the text.
Thus, 17559.26 Hz is the actual sample clock rate for the analog input task.
For Analog Output tasks and Digital I/O:
So, for the example given:
Thus, 17543.86 is the actual sample rate for the analog output or digital task
Programmatic Access in LabVIEW:
You can read the actual sample rate used through DAQmx Timing Property Nodes by selecting Sample Clock » Rate. Below is an example of this in LabVIEW:
Figure 1: Using a DAQmx Timing Property Node to Read Coerced Sample Rate
Collaborate with other users in our discussion forums
A valid service agreement may be required, and support options vary by country.
Thank you for your feedback.
You changed your response to Not Helpful
You changed your response to Helpful